Nov 19, 2018 - 02:00 PM
You will not be able to use the Snap On pressure transducer presets. All readings will be shown as voltage.
For instance on:
Range 1 (0-500psi) the conversion is 1V=100psi.
Range 2 (0-50psi) conversion is 1V=10psi
Range 3 (0-5psi) the conversion is 1V=1psi.
So if you are performing an in-cylinder compression waveform, a reading of 1.20v would be 120 psi.