1. Connect the probe into the COMM and Volt Input. Do not connect to the AMPS input. 2. Set the meter to Millivolts because the probe outputs millivolts/amp. 3. With the AES Generic Low Current Probe the ranges are:
On the meter screen you are really seeing volts, but the probe is measuring amps and displaying the amps on the screen in the form of a voltage reading, scaled per the above. The calibration settings of the AES# 07-60 are hard to understand. We suggest you relabel the settings to make them easier to understand.